Temperature correction factor.
Delivery correction to 60 degrees Fahrenheit.
Meter factor. The meter factor equals the prover quantity at 60
degrees Fahrenheit divided by the meter quantity at 60 degrees
Fahrenheit divided by the meter quantity at 60 degrees Fahrenheit. The
factor is carried out to four (4) decimal places.
Meter certification record. The meter certification record in
conjunction with the meter test results is your proof that the meter has
been properly calibrated. The following items will be included in the
meter certification and recorded to ensure that the meter has been
properly tested and calibrated when required:
Date meter was verified.
Was meter factor within .0025.
Required but Not
If adjustment is required but not performed, the meter factor should
be used to adjust all issues or receipts through the meter.
Use of the meter verification record.
Remarks: Any repairs or adjustments to the meter should be recorded.