(3) Look at meter reading. In order for meter to be within tolerance, the mater should read within the
nearest tenth of a gallon for the prover can capacity.
(4) If meter is within tolerance, continue to use. If meter is not within tolerance discontinue use and
report to organizational maintenance for calibration or repair/replacement.